Easy to use at home, at work or anywhere in the world
Saves clinical time and reduces treatment costs
Reduces the need for clinic appointments
A low cost alternative to residential rehab
Post detox aftercare
Criminal Justice System
The Zenalyser® is a unique dual sensor breathalyser that detects both disulfiram metabolites and alcohol on the breath. It can be used remotely…
ZenaMed is a new British company developing technological solutions that help people abstain from alcohol…
“The combination of taking Antabuse and using the Zenalyser on a daily basis has been an extremely effective method of helping me remain totally abstinent for the following reasons….”
It is important to interpret the data from the Zenalyser correctly. The examples below will help you to understand what the results mean.
A graph of a single 10 second breath sample will be presented on the Run page and at the bottom of the Client Data page.
This is a successful result.
There are lines of 3 colours:
– the green line shows the level of disulfiram metabolite
– the red line shows the level of alcohol
– the horizontal orange line is the “therapeutic threshold”
The green (disulfiram) line should be above the orange line (5 ppm)
Each reading from the Zenalyser is automatically plotted consecutively on the graph at the top of the Client Data page:
This pair of graphs shows a typical run of successful Zenalyser readings over three weeks.
The top graph shows cumulative data.
The lower graph shows the last individual 10 second sample, but other individual samples can be viewed simply by Clicking on the sample line in the upper table.
2. A client drinking alcohol will produce a graph that looks like this:
This is an unsuccessful result.
Both the green line and the red line have gone above the orange threshold line.
The red line indicates that alcohol has been consumed
The green line detects both disulfiram metabolites AND alcohol
This result shows that the client is not abstinent from alcohol and therefore requires quick review or intervention.